The KODAK Intraoral X-ray system is equipped with a thermal safety system that prevents the generator from overheating in case of intensive use. Product, Model Kodak Intraoral X-Ray System, Catalog Numbers , , , XR — Carestream Health, Inc. Made in France Kodak Intraoral X-Ray System Performance. Operational ease. Safety. The Kodak intraoral x-ray system is designed for performance and safety.
|Country:||United Arab Emirates|
|Published (Last):||23 April 2014|
|PDF File Size:||7.44 Mb|
|ePub File Size:||4.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
Live Webinar Stay out of Jail: And, after each exposure, the dose level is displayed for simple dose monitoring. Hi, KV error on a means either a power board failure or a tubehead failure which is the most frequent problem. Timer Emitted dose display User Menu – settings customization Technical Menus — Interface with firmware — 2 counters for exposures — Up to 59 management error codes — Calibrated generator parameters Kodak Intraoral X-ray System — New timer.
Kodak Intraoral X-Ray System | Pre-Owned Dental Inc.
If you wish to download it, please recommend it to your friends in any social system. A revolutionary paradigm shift in dynamic surgical navigation.
It will give you all error codes with counters. The voltage devider must be immersed in oil completely, no air bubbles present in oil! Avoid Coding Errors 12 Feb It includes an intuitive control wheel for easy tooth and program selection, and automatically calculates the optimum exposure time and dose settings. If you find the good value, then check contacts on connecto arm sideplace tube back and check that the resistor reaches the power board by measuring the resistance on the small white connector J6 between black and yellow wire.
Kodak Intraoral X-Ray System – IMAGELEVEL – PDF Catalogs | Technical Documentation
Would systrm thankful if you could send me the tech manual as well. Unless Trophy designed it with certain service lifetime of each unit.
In addition, the timer is pre-programmed to automatically calculate settings—so operating the unit has never been so simple. Then insert the assembly gently inside the metal housing, taking care with the position.
Share buttons are a little bit lower. Equipped with a focal spot of 0. Could you please inform me what I need to do? Faustinio I am also having KV error and the system beeping.
Using PEEK frameworks for full arch implant supported prosthetics. As a part of its mission to always x-eay dentists with only the best solutions for improved patient care, ContacEZ continues to intdaoral, improve and expand its product lines to make them safer and more effective than effort.
It gives KV Error along with bip and on the control panel two green lights turn on. I would recommend you not dismantle the ceramic resistor. Sign In or Register.
Used & Refurbished Dental Equipment
As part of its growth plans and in order to better serve its customers, ASI has announced that it is adding local sales representatives to cover designated territories throughout the U. Back to news USA. Content you may be interested in: December 13, New law seeks to improve access to oral health care for underserved patients.
In this case there’s no high voltage and you have KV error. Thanks, I’ll try to get the koddak from Start by measuring the 13 Kohm feedback resistor located inside the housing with Ohmmeter between ground and yellow small wire on the tubehead connector once removed from the arm.
For next procedure for self-calibration: Advertise Increase Your Brand Awareness. Easy, precise positioning With its stable arm and lightweight tube head, the Kodak system ensures perfect stability during the exam and effortless movement.
Wait the end of the self-calibration sequence for about 15 sec. Kodal 15, Business ASI is adding independent sales representatives As part of its growth plans and in order to better serve its customers, ASI has announced that it is adding local sales representatives to cover designated territories throughout the U.
Is it possible if I x-rah use a regular square ceramic resistor 5 watt? The soldering on the resistor gets loose and this is the reason why so many of you get error December 7, Meeting review: Air is a reason of fail in repair.